Polystyrene Sonication Tubes
enhanced energy transfer for more efficient sonication shearing
Because polystyrene transfers sonic wave energy much more efficiently than polypropylene, Active Motif highly recommends that its Polystyrene Sonication Tubes be used with its EpiShear™ Multi-Sample Sonicator. Use of polypropylene tubes in indirect sonication has been shown to reduce the extent to which the chromatin is sheared (see the Tube Comparison tab below).
Figure 1: Polystyrene Sonication Tubes.
|Sonication Tubes / Caps, 1.5 ml polystyrene||200 tubes||53071||$75||Buy Now|
|1000 tubes||53072||$300||Buy Now|
Improved shearing results using polystyrene tubes
The EpiShear Multi-Sample Sonicator was used to shear chromatin from 3T3 cell samples in both polystyrene and polypropylene tubes. Because cup horn sonicators use an indirect shearing method that requires the sonic wave energy to pass through the water bath and the wall of the tube, use of polystyrene tubes enables more efficient shearing, with far less high-molecular weight material remaining over after sonication (Figure 2). Therefore, Active Motif highly recommends use of its Polystyrene Sonication Tubes with its EpiShear Multi-Sample Sonicator, as well as with other manufacturer's cup horn sonicators.
Figure 2: Chromatin shearing in polystyrene versus polypropylene tubes.
One or five bags containing 200 polystyrene sonication tubes (1.5 ml, non-sterile) and 200 polyethylene caps per bag.